We can build the three-dimensional structure model based on the Gambit software and achieve the distribution of flow field in the pipe and reflux flow condition at the position of transducer in regard to the real position of transducer according to the Fluent software. Under the framework, define the reflux length based on the distance of reflux along the channel and evaluate the effect of reflux on flow field. Then we can correct the power factor with the transmission speed difference method in the ideal condition and obtain the matching expression of power correction factor according to the practice model. In the end, analyze the simulation experience and produce the sample table based on the proposed model. The comparative analysis of test results and simulation results demonstrates the validity and feasibility of the proposed simulation method. The research in this paper will lay a foundation for further study on the optimization of ultrasonic flowmeter, enhance the measurement precision, and extend the application of engineering.
"The Flow Field Analysis and Flow Calculation of Ultrasonic Flowmeter Based on the Fluent Software." Abstr. Appl. Anal. 2014 (SI02) 1 - 8, 2014. https://doi.org/10.1155/2014/528602